Too bad you're not a member of the forum. We could send you the Provider's manual for the machine, which has all of this information......and more.
However, this is from that manual concerning the Device Alerts which may apply to your situation.
Den5.1 Device Alerts
The device provides two alert levels, high and medium priority.
• High Priority – These alerts require immediate operator response. The alert signal consists
of a high priority sound, which is a continuous two-beep pattern (indicated in the following
table as: • • • •). Additionally, the backlights on the buttons provide a high priority
flashing pattern consisting of a continuous, bright-to-off, two-flash pattern (indicated in the
following table as: ◊◊ ◊◊ ◊◊).
• Medium Priority – These alerts require prompt operator response. The alert signal consists of
a medium priority sound, which is a continuous one-beep pattern (indicated in the following
table as: • •). Additionally, the backlights on the buttons provide a medium priority
flashing pattern consisting of a continuous, bright-to-dim, one-flash pattern (indicated in the
following table as: ◊ ◊ ◊).
System ErrorAudible Indicator:
• • • • Visual Indicator:
Backlights: ◊◊ ◊◊ ◊◊
Screen displays the following symbol, indicating that service is required. (triangle with exclamation mark)Device Action:
The device enters the “Safe state” in which the device power remains on, but the airflow is disabled.Possible Cause:
Device failure Patient Action:
Press any of the display screen buttons to silence the alert. Remove the power supply cord from the device to remove power. Plug the cord back into the device’s power inlet to restore power. If the alert continues to occur, contact your home care provider.Provider Action:
If you have another power supply available, try changing the power supply and reapplying power to the device. If this resolves the problem, send the defective power supply to an authorized service representative or Respironics for service.
If trying another power supply does not correct the problem, or if you do not have a spare power supply, send the device and power supply to Respironics or an authorized service center for service. Please have the serial number ready when you call.
(4) REMstar Autos w/C-Flex & (6) REMstar Pro 2 CPAPs w/C-Flex - Pressure Setting = 12 cm.
"Passover" Humidification - ResMed Ultra Mirage FF - Encore Pro w/Card Reader & MyEncore software - Chiroflow pillow
User since 05/14/05