|Mask: Swift™ LT Nasal Pillow CPAP Mask with Headgear|
|Additional Comments: Resmed Aircurve 10 ASV, ASVAUTO Min Epap 8.4, Max Epap 10.6, PS 1.6-11.4|
klv329 wrote:It looks like the honeycombed 'airflow baffle' in step number 8 is reversed in direction in the elite victim relative to PR's linked pictures of a v-auto victim. But somehow I doubt that it matters.....
Naceur wrote:Thank you for this advice: my CPAP can not start; when I press the start button: the screen lights up but it seems that loop without passing the startup stage ..that you advise me to do ??
doitrightkid wrote:WOW Thank you! You cant know how much this helped me out of a jam my new insurance company cant help whats so ever , I'm having some difficult time locating a new filter housing or holder for this . thank you again. Phil
JGaspar wrote:Hi everybody!
I did the disassembly, and found my S9 loose on expiratory after assembly, but I tested on barometer and the pressure is OK. I disabled EPR too. Anyone ever noticed the same? Any help?
Users browsing this forum: No registered users and 101 guests